Many switches in the autoclave equipment are controlled by handles, which are mainly used for convenience and more direct use of the handles. In addition, there are some handwheels, buttons and other operations. Many switches for mechanical transmission are used. Handle or hand wheel, and electric operation and other aspects use buttons for related operations. Today we will introduce the switch operation during the production of autoclave.
1. The operator must be familiar with the instruction manual, be familiar with the position of the operating handle, handwheel, and button, and operate in the prescribed order. 2. Operators of non-autoclaved tank equipment shall not arbitrarily move the handle, handwheel or push buttons and other related mechanisms. 3. When the product enters the tank or the end of the work, the operating procedures to close the door: 3.1 Press and hold the "rail bridge lift" button in the on-site button box, and the rail bridge will stop automatically when it is lifted into place (confirm that the "bridge lift into place" indicator is on). 3.2. Press and hold the "Tank Close" button on the on-site button box, the tank door will stop automatically when it is closed in place (make sure that the "door closed in position" indicator is on). 3.3. Press and hold the "Tank Tightening" button of the on-site button box, and the tank door will stop automatically when it is turned tightly (make sure that the "door tightly in place" indicator is on). 3.4. Move the handle of the "safety device" upwards. The handle is locked in place. (Make sure that the "safety in place" indicator is on).
Note: (The relevant operations may vary slightly between different products but the general idea is the same).
4. When the product is out of the tank or the work is completed, the operating procedures for opening the door are required: 4.1. After the product is cured, carefully check the status of each safety interlock device. 4.2. When the pressure in the tank reaches zero pressure, turn the "safety device" handle "down" (due to different equipment, the handle may be turned in the opposite direction, see the actual product) in place to open the control, and let the remaining gas in the control gas pipeline After the gas is exhausted, the remaining pressure in the tank is zero (the "zero pressure indicator" light on the operation console is on) before proceeding. 4.3. Press and hold the "Release Tank Door" button on the on-site button box, the tank door begins to loosen, and the tank door will automatically stop for 5 seconds when it turns to the safety tooth position, waiting for the remaining gas in the tank to be released. Continue to press the "Door to Loose" button, and the door will stop automatically when it is turned into place. (Confirm that the "door loose in place" indicator is on). The above is the opening and closing operation of the autoclave produced by our factory brought to you by Zhucheng Xintai. In fact, the general way of operation of the autoclave about this piece is the same. The handle positions of different manufacturers are the same. And labeling may be different, but many places are common. If you do n’t know anything about this, you can always consult our Zhucheng Xintai New Materials Co., Ltd., our company specializes in producing all kinds of heat. We can customize the pressure tank equipment from small to large. You are welcome to call us for consultation.